Nov 07, 2016
It is important that Ultra Low Temperature Freezers be placed in your laboratory appropriately to ensure maximum efficiency, safety and longer equipment life.
Here are some pointers that you can follow in choosing the right spot for your ULT freezer.
Position your unit in a firm and level floor in an indoor environment.
Allow at least 150 millimeters (6 inches) or open space on all sides, top and rear. Sufficient ventilation must be considered to prevent high ambient temperature.
Avoid placing your ULT freezer near any heat source as this may increase energy consumption.
The recommended operating temperature is 25°C with a relative humidity of at least 90% which is non-condensing.
Watch this video for more information:
For more videos and information on Esco products, visit our YouTube channel at www.youtube.com/EscoGlobal
Esco Technologies, Inc.
903 Sheehy Drive, Suite F, Horsham, PA 19044, USA
Toll-Free USA and Canada: 1-877-479-3726
215-441-9661 | 484-698-7757
2512 Metroprolitan Dr, Suite 120-B
Feasterville-Trevose, PA 19053-6738
+1 215-322-2155 - Ext 1101